We can offer 2Mhz,4Mhz,6Mhz TCD Doppler Sensor .The shape of shell can be customized based on your requirements.
Application:This probe is used for craniocervical doppler flowmeter,it is good anti-interference ability and high sensitivity,which
has been integrated with many TCD equipment manufacturer.
The Doppler blood measuring device we developed can equip with 2mhz, 4mhz, 8mhz blood flow sensor to measure the different depth blood flow rate. The output interface is USB port, and input the power is standard PC power. There could offer some sort of raw data or the calculated data as Vp; Vm, Vd, PI, RI, SD, Hr. Besides, we could offer the output as bitmap spectrum when connect with your laptop or PC directly.
4MHz Craniocervical Doppler probe:
IPX7 4000V（against ground）
Length of Cable:
For brain blood flow analysis
For the carotid artery blood flow analysis
For superficial vascular blood flow analysis
These parameters for most patients (age above 20 years old) are mainly based on the detection of vascular anatomy depth.
h. Left or right eye arterial general depth of 45mm (power must be less than 250%, otherwise injury eyeballs)
General Power = 400% except the ophthalmic artery.
Change principles: appropriate increase or decrease with the detection depth.
The gain is usually about 30dB.
Change principles: appropriate increase or decrease with the detected signal strength. Without changing the general set according to user habits.
Generally it is about 12mm.
Change the principle: If taken large, easily detected signal, but the vascular low resolution, and vice versa more difficult detection signal, but the vascular high resolution. Generally for deeper blood vessels should be bigger, lighter is smaller.
Change the principle: limited detection depth, directly determine the flow rate range. Its value with Depth and Sample volume. Since receive data, first of all deal with the relationship of these parameters do spectrum window must clearly can not do otherwise, the flow rate scale. All these need to be changed at any time with the detection depth.
Change principles: interface design, spectrum window can be longer or shorter. Spectrum window is the longer, the longer time. For a fixed interface, this parameter set is no longer changed. For example, for the 256-point FFT FIG longitudinal width must point pixel 256, while the horizontal generally taken per 80-100 points / S is more appropriate. Need to be doubled for a 512-point FFT lateral more appropriate, or spectrogram sawtooth will seem too high, sharp or too horizontal long influence doctors interpret blueprints. Multi-integer convenient. For example: 4S or 5S, 6S like.